Mechanical Ventilation- Pressure Support Ventilation Graphs
This is the information from a pressure supported breath. We have the pressure generated at the top, the flow generated in the middle and the subsequent volume at the bottom.
There are a number of important points to note. If we look at the pressure graph, we can see a slight dip below the base line before each breath, where each red arrow is pointing. This is the negative pressure created by the patient as they try to initiate a breath.
Therefore the breath will always be patient triggered in pressure support ventilation.
Each breath, after initiation, will then be pressure limited as you can see by the plateau on each breath. We set that pressure when we decide how much pressure support the patient needs.
If we look at the flow curves in the middle section we can see the difference between pressure controlled and pressure support ventilation. In pressure support ventilation the breath is cycled by flow.
So as the patient takes a breath in they generate an inspiratory flow which will peak at point 1 on this diagram and will then decline down the point two, which is when the ventilator will then cycle to the next breath. In pressure controlled ventilation, where we set a rate, the breath is time cycled.
The consequence of this on the volume curve is that we don't see an inspiratory pause. As soon as the ventilator detects a reduction in the flow rate it triggers to expiration. The breath is not held for a period of time as it would be in pressure controlled ventilation.
- Peak and Plateau Pressure
- AC versus SIMV mode
- Positive End Expiratory Pressure (PEEP)
- Increase the rate or tidal volume?
- Phases of a breath- I:E ratio and cycle time
- Ventilation screen- what do those numbers mean?
- Pressure Support
- Modes of ventilation I
- Modes of ventilation II
- Physiologic effects
- Physiologic goals
- How do I describe how my patients ventilation?
- Trigger, Limit and Cycle
- Pressure support ventilation graphs
- ARDS and Proning
- 6 ways to be better with Bag-Valve-Mask
- Phase Variables
- Airway Pressure Release Ventilation (APRV)
- Pressure Volume Loop
- Lung compliance in volume controlled ventilation
- Pressure/Volume/Flow graphs
- A-a gradient
- Goals and Indications
- Anatomy of the Endotracheal Tube
- Lung Compliance
- Ventilation/Perfusion V/Q matching
- Ventilator Induced Lung Injury (VILI)
- Ventilator Associated Pneumonia (VAP)
- Phase variables...again...
Guidelines for the management of tracheal intubation in critically ill adults
Having read the guidelines I made these infographics. They are FREE. Just let me know your email address and they will be sent to you.
Copyright 2019, Critical Care Practitioner - Disclaimer